Comparison of Spectral Reflectance and Stylus Profilometry
Stylus profilometry is an easy-to-understand mechanical technique for measuring film thickness. It works by monitoring the height of a stylus as it drags across a step in the film (see figure to right). The primary advantage of stylus profilometry is that it can be used on all solid films, including opaque materials such as thick metal films. (For information about our low-cost optical profilometer, click here.)Get Spectral Reflectance Guide
There are, however, disadvantages to stylus profilometry. Most importantly, a sample with a step must be formed to measure film thickness, and film steps thus formed are often not perfect (see figure). This, combined with calibration error and mechanical drift, can often lead to significant errors in measured thickness.
In contrast, Spectral Reflectance (SR) is a non-contact technique that requires no sample preparation to measure film thickness. Light that is reflected from the film is analyzed in about one second to determine both a film’s thickness and refractive index. Multiple-layer films can also be measured with SR.
A summary of the main advantages of each technique is listed below.
|SPECTRAL REFLECTANCE||STYLUS PROFILOMETRY|
|Thickness Measurement Range:||
1nm - 3mm (non-metals)
0.5nm - 50nm (metals)*
2nm - 0.5mm (hard materials)
|Measure Multiple Layers:||Yes||No|
|Measure Refractive Index:||Yes||No|
|Repeatability (500nm SiO2):||0.1nm||0.5nm|
|Measurement Speed:||<1 second||~5 - 30 seconds|
|Sample Prep Required:||No||Yes|
|Price of Basic System:||~$10K||~$40K|
Contact Us Now
Contact Us Now
Learn about the importance of thin films with our animated short film, or see how easy it is to set up and use a Filmetrics system.More…